Adapt symptoms: suit for trigeminal neuralgia and primary Intervertebral Disc pain that is ineffective after regular conservative treatment.
Reusable RF electrode probe, Single Unified electrode RF cannula, Single use RF cannula
We're here to help:
Easy ways to get the answers you need.
Working voltage:220±10% 50Hz±1Hz.
Have Auto control and manual control mode,
damage temperature and time are adjustable.
Temperature setting scope: 50°C-95°C.
Temperature measure scope: 25°C-95°C
Pulse RF width control mode: 10-50 ms
Pulse RF frequency mode: 1-8 Hz.
Could change with continuous RF freely
Stimulation pulse width: 1-3ms
Stimulation pulse amplitude: 0.1-5.0V continuous adjustable
Stimulation pulse frequency: 1Hz-90Hz
Therapy temperature scope more than 5°C, lesion generator will stop working.
Please fill in your procurement needs and contact information